DEFIBRILLATOR (Biphasic) Technical specifications 1) Defibrillator type: Manual, Synchronized, Asynchronized 2) Energy accuracy: ∠+1% 3) Energy selection: 0, 3, 5, 7, 10, 20, 30, 50, 100, 200, 300, 360 joules (nominal at a resistance of 50) 4) Charging time: Maximum 10 seconds to 360 Joules 5) Standard Adult/Pediatric Paddles: Reusable external adult paddles (pediatric paddles integrated)
Power requirements (a) AC Power supply 100V~240V, 50Hz/60 Hz (b) Vehicle Voltage (Option) DC 12V
Internal Battery Backup (1) Battery type: Rechargeable lead-acid battery (12V 2.0Ah X2) (2) Charging time: Time-Minimum of 4 hours for full charging (3) Battery charging: During AC Power Supply operation of the device. (4) Charging /Low Battery Indication: Red LED (5) Battery Capacity: 100% of the battery will provide 120-min monitoring; and 35 discharges at 360 joules.
Display LCD Dimensions 7-inch Type High resolution true-color graphic Resolution 480*234
ECG ECG lead: I, II, III, AVR, AVL, AVF, V1~V6 S-T segment analysis: -2.0~2.0mv Arrhythmia analysis: Yes Protection: With stand 4000vAc/50Hz voltage in isolation and work against electrosurgical knife and defibrillation